09-03-2021 02:26 PM
We are using BME680 sensor in our data loggers. I noticed, that it has to recalibrate quite often. The device lays in one place, in a typical room and there are no extreme VOC changes. The chart below shows IAQ index measured by the sensor. Parts of the chart mared with red are the moments when the sensor was re-calibrating. Questions:
09-06-2021 08:20 AM
Is the corresponding IAQ accuracy of the data in the red part(Not calibrated) 0?
09-06-2021 09:55 AM
on the chart:
If the snesor returns "0", we do not present it on the chart at all. So the main uestion is, if the users can trust the data in red (1) and treat it as realible readings.
09-10-2021 10:16 AM
@BSTRobin, do you have any updates on this? Can we treat the data with calibraiton level 1 as reliable?
09-29-2021 03:07 PM
could you please confirm that we can trust the data with calibraiton levles 1,2 and 3?