Before we repoduce your case, we would like to know what is the HW environment you used? APP2.0+BNO055 shuttle?
And for your attachement picture, please mark BNO055 axis in the picture.
BTSrobin, nice to hear from you. Ill get back to you with these answers.
I found a flaw in my magnetometer calibration code that more-or-less solved the problem i was refering to here.
I posted about another probelm i noticed where the system calibration CS value is unstable, while the CA, CG, and CM values are all stable at 3. I can see that when the CS value deviates from 3, so does our heading. What could be causing this?
In order to further analyze this problem, we should know what is the HW environment you used? APP2.0+BNO055 shuttle?
And for your attachement picture, please mark BNO055 axis in the picture and upload it again.
We will also reproduce your feedback that CS value is unstable.
Thank you for your cooperation.
Thank you for your help. Regarding our HW environement. We are not using the APP2.0 or the BNO shuttle. Our BNO is mounted PCB that communictaes with a custom central baord.
Here are the relabaled pictures of the axis of the BNO at start of the tests.
Thank you for your continued help. It is much appreciated.
Thank for your feedback.
We haven't reproduced your issue with APP2.0+BNO055 shuttle boards now.
Do you have APP2.0+BNO055 shuttle boards on your hand? If yes, you could try to reproduce it and let me know how to reproduce it.